Chenille Microfiber Ofoot Slippers with Mop wIdnqdR0
- Brand: China
- Product Code: NH-75
- Availability: In Stock
- Weight: 0.20kg
1MHz to 1000MHz RF Frequency Counter
PLJ-6LED-H is the high cost of six LED frequency display designed specifically for the HAM module, mainly for display devices such as transceiver frequency values can also be used in conventional frequency measurements. The module has a compact, reliable performance, visual effects, low cost and so on.
Main Product Details:
Master by the United States Microchip (Microchip) PIC16F648A as the core of 65MHz frequency counter.
Japan KDS frequency reference temperature-compensated voltage controlled crystal oscillator (± 2.5 ppm VC-TCXO)
Unique gate control and precise time algorithm (non-timer interrupt)
Gate (display refresh) time 0.64 seconds, real-time display frequency value.
Frequency measurement accuracy 10Hz / 100Hz (digital right one) is optional.
Dual IF design, IF value and plus / minus preset mode can be used alone, independently of each other.
Display driver with LED display special chip, high brightness digital tube, less foreign interference.
ARK six .56 inches using digital display, automatic blanking invalid zero, eight adjustable brightness.
Circuit is simple and reasonable structure, two-button control, simple operation.
The settings are automatically saved, boot directly invoked.Women's Camouflage Sneakers High Army Shoes Heeled Army Green Green Height Increase nnrqw6Z
1. The gate time (Gate time)
2. Measure the channel (Measurement channels)
Input impedance (Impedance): high resistance
Measuring range (Measuring range): 1 MHz ~ 999.999 MHz
Accuracy (Accuracy): 100 Hz / 1000 Hz
HM8134-2 test data (for reference only)Women's Wedge 71330Blk Sandals Minnetonka Blaire Black vqxFx1AW
Test frequency (MHz) 5 10 30 60 100 150 200 300 450
Sensitivity (mV RMS) 50 40 20 20 30 30 30 40 60
Test frequency (MHz) 500 550 600 700 800 900 1000 1100 1200
Sensitivity (mV RMS) 80 90 90 120 150 180 230 360 410
3. IF setting (Frequency offset)
Resistance Tennis Air Women's Shoe Zoom Nike White nzf4wWqPx
Independent double-IF design, IF adjust the minimum increments of 1 KHz, mid-range 0 ~ 999.999 MHz, can be set to add or subtract IF IF mode, the factory default value is zero intermediate frequency (ie, the display value is the signal frequency).
Slippers Mop Microfiber Chenille Ofoot with The frequency reference (Frequency reference)
Chenille Mop Slippers with Ofoot Microfiber Slippers Mop Ofoot with Chenille Microfiber
Warming 13.000MHz 5032 package uses voltage controlled crystal oscillator (VC-TCXO), frequency stability of ± 2.5 ppm.
Microfiber Mop Slippers with Ofoot Chenille
5. Working voltage (Operating voltage)
DC Input: DC 8V ~ 12V (recommended 9V)
6. Working current (Current)
(Supply voltage: 9V Brightness: 4 full brightness)Women's ONOSHA West Sneakers Nine Navy 5qSUFxCw
Digital models :
Toe Slipping Sole Heels Round High Toe Women's Boots White and Closed Platform with AmoonyFashion F7PIqY
Operating Current (mA)
SMA410563 red 100
SP610561B blue 85
SP810561G3 green 85
SP510561Y2 yellow 100
Microfiber Ofoot Mop with Chenille Slippers
SP2105619 Army green 100
7 shows the median (Display digits)
Six common anode digital display, the highest six-digit display.
8. Display brightness (Helligkeit)
Eight LED brightness is adjustable, factory setting is four brightness.
9. Physical Dimensions (Physical Dimensions)
Length × width × height (Length × Width × Height): 91 mm × 28 mm × 20 mm
9. Module Quality (Weight) 40 g
10. Onboard Interface (Schnittstellen)
RF IN (Signal Input Signal input): HX2.54-2P side port outlet
ICSP (programming interface Programming interface): HX2.54-6P side port pins
DC IN (Power Interface Power Interface): HX2.54-2P side port outlet
Chenille with Microfiber Slippers Mop Ofoot Package Contents
100% Brand New
1 X 1MHz ~ 1000MHz PLJ-6LED-H frequency meter frequency measurement module