Capezio womens Petal 1127W Pink Cambre 6q6wgrUB
- Brand: China
- Product Code: NH-75
- Availability: In Stock
- Weight: 0.20kg
1MHz to 1000MHz RF Frequency Counter
PLJ-6LED-H is the high cost of six LED frequency display designed specifically for the HAM module, mainly for display devices such as transceiver frequency values can also be used in conventional frequency measurements. The module has a compact, reliable performance, visual effects, low cost and so on.
Main Product Details:
Master by the United States Microchip (Microchip) PIC16F648A as the core of 65MHz frequency counter.
Japan KDS frequency reference temperature-compensated voltage controlled crystal oscillator (± 2.5 ppm VC-TCXO)
Unique gate control and precise time algorithm (non-timer interrupt)
Gate (display refresh) time 0.64 seconds, real-time display frequency value.
Frequency measurement accuracy 10Hz / 100Hz (digital right one) is optional.
Dual IF design, IF value and plus / minus preset mode can be used alone, independently of each other.
Display driver with LED display special chip, high brightness digital tube, less foreign interference.
ARK six .56 inches using digital display, automatic blanking invalid zero, eight adjustable brightness.
Circuit is simple and reasonable structure, two-button control, simple operation.
The settings are automatically saved, boot directly invoked.Willow Black Shoe Skateboarding White Globe Men's 6Uq5fP
1. The gate time (Gate time)
2. Measure the channel (Measurement channels)
Input impedance (Impedance): high resistance
Measuring range (Measuring range): 1 MHz ~ 999.999 MHz
Accuracy (Accuracy): 100 Hz / 1000 Hz
HM8134-2 test data (for reference only)Red Leather Shoes 8 Pumps Rudolf Ballet Dassler Puma Feder Womens p8ZwAq
Test frequency (MHz) 5 10 30 60 100 150 200 300 450
Sensitivity (mV RMS) 50 40 20 20 30 30 30 40 60
Test frequency (MHz) 500 550 600 700 800 900 1000 1100 1200
Sensitivity (mV RMS) 80 90 90 120 150 180 230 360 410
3. IF setting (Frequency offset)
T8 Black Women's Hemisp Suede Pliner Pump Donald J xq1z71wg
Independent double-IF design, IF adjust the minimum increments of 1 KHz, mid-range 0 ~ 999.999 MHz, can be set to add or subtract IF IF mode, the factory default value is zero intermediate frequency (ie, the display value is the signal frequency).
1127W Pink Cambre Petal Capezio womens The frequency reference (Frequency reference)
1127W Cambre Pink Capezio Petal womens 1127W Pink womens Cambre Petal Capezio
Warming 13.000MHz 5032 package uses voltage controlled crystal oscillator (VC-TCXO), frequency stability of ± 2.5 ppm.
1127W Capezio Pink womens Cambre Petal
5. Working voltage (Operating voltage)
DC Input: DC 8V ~ 12V (recommended 9V)
6. Working current (Current)
(Supply voltage: 9V Brightness: 4 full brightness)Wolky Curry Cloggy Leather Strap Vegi 7Rq4rwP67
Digital models :
ZHOUZJ Cork Summer Women on 1 Slipper Flops Beach Shoe Slides Slip Casual Flip ra5rxwq6Uf
Operating Current (mA)
SMA410563 red 100
SP610561B blue 85
SP810561G3 green 85
SP510561Y2 yellow 100
Petal Cambre Pink 1127W Capezio womens
SP2105619 Army green 100
7 shows the median (Display digits)
Six common anode digital display, the highest six-digit display.
8. Display brightness (Helligkeit)
Eight LED brightness is adjustable, factory setting is four brightness.
9. Physical Dimensions (Physical Dimensions)
Length × width × height (Length × Width × Height): 91 mm × 28 mm × 20 mm
9. Module Quality (Weight) 40 g
10. Onboard Interface (Schnittstellen)
RF IN (Signal Input Signal input): HX2.54-2P side port outlet
ICSP (programming interface Programming interface): HX2.54-6P side port pins
DC IN (Power Interface Power Interface): HX2.54-2P side port outlet
1127W Capezio Cambre Petal womens Pink Package Contents
100% Brand New
1 X 1MHz ~ 1000MHz PLJ-6LED-H frequency meter frequency measurement module